The MWTC-REC wireless receivers support up to 48 MWTC miniature sensor / transmitters. The MWTC-REC1 connects to a PC via a 6 foot USB cable while the MWTC-REC5 is a dongle the size of a flash drive that plugs directly into the USB port of a PC. Both receivers allow you to display, chart, and save data from multiple MWTC sensor / transmitters in real-time using the free downloadable TC Central software. The MWTC product line is only available in North America.
Would this device be compatible with any type of probe . we are currently loooking for a device to measure compost temperature and humidity via some kind of probe.
Thank you !
Asked by: Marc Andre Lauzon
As long as it is one of the following thermocouple types: J, K, T, E, R, S, B, C or N
Also the probe will need either a standard round prong or mini blade type connector attached at the end so it can connect to the transmitter.
Answered by: JA Engineer
Date published: 2019-08-28
We are interesting to purchase the product but we need some clarifications for this products, If the equipment is 24/7 working (Recorded) or only particular time working (Auto power off without disturbing the equipment? please advice.
Asked by: Urjita Electronics
When you are setting up the logger you can select either to stop on Memory Full or Memory Rollover for Continuous Recording.
Answered by: Application Engineer
Date published: 2019-07-22
I have the HH-309A temp mapping tool, it was working while I was running on windows 7 but not that I am running windows 10 the tool no longer shows a connection to the computer. Is there another program that I need to download?
Asked by: Scott White
The problem is the driver. Not the program.
I have the HH-309A as well and the communication to the unit stopped working with Windows 10.
Luckily Windows 10 identifies my USB to serial cable (labeled USB-300 by Omega) as:
- CP2102 USB to UART serial connection.
But the provided driver by Microsoft did not work... nor the original driver by Omega.
I found the Silicon Labs CP210x driver to USB connection in this site: https://www.pololu.com/docs/0J7/all#2
I installed the X64 bit version of it and it now shows in the Windows Device Manager under Port (COM & LPT)
as: Silicon Labs CP210x USB to UART Bridge (COM1)
I re-installed the SE309 Software from the original mini-CD disk that came with the HH-309A unit and everything
is now working perfect.
Home this helped.
Answered by: AEC007
Date published: 2019-07-19
Hi we just buy a bunch of these monitors but we don't have cd-room; can we download the software?
Asked by: Efcamb
Yes, please visit use the link below to download the software, thank you.
Answered by: Jane
Date published: 2019-04-12
What is the fastest sampling rate?
Asked by: Simon
Thank you for connecting with Omega. 1 sample per 2 secs is the fastest sampling rate which is depend on the RF reader.