Le CL3515R est un calibrateur/thermomètre avec un affichage rétroéclairé à 4 digits. Il a été conçu pour mesurer et simuler une large plage de types de thermocouples: K/J/T/E/R/S/N/L/U/B/C. Il y a aussi une plage de tension de sortie de -25mV à + 75mV. Le thermomètre présente une double entrée thermocouple et un offset thermocouple ajustable.
CARACTERISTIQUES Thermocouple (Résolution) plage de mesure*: K: (0,1°) -200 à 1372°C J: (0,1°) -210 à 1200°C T: (0,1°) -250 à 400°C E: (0,1°) -250 à 1000°C R: (1°) 0 à 1767°C S: (1°) 0 à 1767°C N: (0,1°) -200 à 1300°C L: (0,1°) -200 à 900°C U: (0,1°) -200 à 600°C B: (1°) 600 à 1820°C C: (1°) 0 à 2316°C * Selon la température standard ITS-90. Précision: Thermocouple types K/J/T/E/L/U: ±(0,05% lec + 0,5°C) -50 à 1372°C ±(0,05% lec + 1,0°C) -50 à -250°C Thermocouple type N: ±(0,05% lec + 1,0°C) -50 à 0°C ±(0,05% lec + 0,5°C) 0 à 1300°C Thermocouple types R/S/B/C: ±(0,05% lec + 2°C) 0 à 1767°C Plage de simulation de thermocouple: Résolution: 0,1° (1° pour R/S/B/C) Précision*: ±(0,3°C + 10 µV) * Précision est spécifiée pour les températures de fonctionnement de 18 à 28°C pedant 1 an, ne comprend pas des erreurs du thermocouple. Plage mV : -25 à 75 mV Résolution: 10 µV Précision: ±0,025% + 1 digit Coefficient de température: 0,1 fois la spécification de précision par °C de 0 à 18°C et 28 à 50°C Affichage: 4 digits, rétroéclairé avec lecture maxi de 19999 Connecteur entrée: prise thermocouple miniature (SMP) Alimentation: batterie 9V (comprise) Autonomie batterie: 17,5 heures, nominale avec indicateur batterie faible Dimensions: H 192 x L 91 x D 52,5 mm Poids: 320 g Plages de fonctionnement ambiantes environnementales: 0 à 50°C, <80% HR Plages de stockage: -20 à 60°C, <70% HR
Rated 2 out of
OK Quality 2 channel calibrated unitWe bought this because it is calibrated. Seems susceptible to EMI. The HVAC system combined with other adjacent CE compliant test equipment cause this to fail the self test. Other thermometers and test equipment do not fail under the same conditions. It is good that it reports the errors, but it would be nice if it was more robust. Good tech support to identify EMI as the cause.
Date published: 2017-06-27
Rated 1 out of
Facilities guy from
Does not work with Windows 10This product will not work with Windows 10 and no relief in sight...
Date published: 2018-12-06
Rated 5 out of
Impressive AccuracyI have purchased 3 of these over the past couple of years for training customers on properly performing SAT's and they work great. They are every bi as accurate as $2,000 calibrators we have used in the past and they handle a wide range of thermocouples.
Date published: 2017-10-10
Rated 1 out of
Rarely in specificationAlways having to adjust and issuing out of tolerances. Get what you pay for, buy a Fluke.
Date published: 2019-10-08
Rated 4 out of
Meets my needsI still have a lot to learn about all the capabilities of this equipment, but it works for my calibration needs. I would like to see more documentation on the capabilities and how to use them. What I have seen thus far is a bit lacking for me. I haven't had the opportunity to play with it outside of performing my calibrations. However, it is great for my calibration needs
Does Omega offer a calibration/re-certification service for the CL3515R, and what is the suggested frequency of recalibration?
Asked by: CalibratorCL3515R
The CL3515R can be re-calibrated and it is recommended it be done once per year. This can be set up by the Customer Service group when you are ready to move forward.
Answered by: ANB ENG
Date published: 2020-06-17
Can I communicate with the unit through USB in my own program instead of the windows program. I am trying to use it under Mac so would like to open it as a serial port to control it. Is there any documents of Remote Programming Commands for the CL3515R?
Asked by: Apple
We do have command protocol for the CL3515R. Please email Temp@omega.com to request copy.
Answered by: AENorwalk
Date published: 2020-04-03
I am having an issue getting the software to see the calibrator. When I plug it in and try to select a port it does not pull up an option to click anything. Any suggestions? Also is there any more information on how to calibrate than what's in the manual?
Asked by: Pyro
Please make sure you are selecting Run As Administrator to install the USB driver. Check the Device Manager to confirm the USB driver is correctly loaded and Windows assign a port number.
Answered by: NJENG
Date published: 2020-02-12
Since most modern computers don't have disc drives, where can I download the software? I don't see it on the website.
Asked by: Some guy
Scroll to the bottom of the page. Under "Quick Links", there is a section called "Software". That will take you to Omega's web page containing all of their software. For this specific item, click on "Temperature", then find this model number on the FTP site.
Answered by: Anonymous
Date published: 2020-01-23
Does this unit come with the optical to USB interface cable, if not what is the P/N for the cable?
Asked by: Stroz
Thanks for connecting with Omega. The product manager has confirmed that this product does come with a USB cable.