Specifications Range: 0 to 200,000 lux; 0 to 18,580 fc Accuracy: ±3% rdg + 10 digits (based on a light source 2856K) Resolution: 1 lux (0 to 19999), 10 lux (20,000 to 200,000), 1 fc (0 to 18,580) Units: lux/fc Operating Temperature: 0 to 50°C (32 to 122°F) Typical Light Intensity Ranges (lux): Design/General Office: 1500 to 2000 Meeting Room: 300 to 750 Precision Assembly: 1500 to 3000 R&D Department: 750 to 1500 Packing/Measurement/Hallway: 300 to 750 Power: 2 “AAA” batteries (included) Battery Life: 180 hours typical with alkaline batteries Dimensions: 24.7 x 50.9 x 132.9 mm (0.98 x 2.0 x 5.2") Weight: 116 g (4.0 oz) Comes complete with batteries, carrying strap, and protection cap.
Optical RMP Meter - HHC250
Specifications Range 100.0 to 99999 Accuracy: ±0.02% of rdg + 1 digit Resolution: 0.1 rpm: 100.1 to 19999.9 rpm 0.1 rps: 1.7 to 333.3 rps 1 rpm: 20,000 to 99,999 1 rps: 334 to 1666 rps Units: rpm/rps Operating Temperature: 0 to 50°C (32 to 122°F) Power: 2 “AAA” batteries (included) Battery Life: 30 hours typical with alkaline batteries Dimensions: 24.7 x 50.9 x 132.9 mm (0.98 x 2.0 x 5.2") Weight: 110 g (3.9 oz) Comes complete with batteries, carrying strap, protection cap, and reflective tape (5 each).
Rated 1 out of
Nice size and design: however doesn't work Bio-MedLiked the size and overall design. Found the battery door hard to open. This was purchased for a Bio-Med shop. Tested unit on centrifuges that you tach through a window. We have Hamilton Bell V6500 centrifuges tinted plastic impossible to get a reading. Then tried on a Beckman Coulter clear plastic window very hard to get reading knowing what I should get, this model display tach is very accurate. Very disappointed it is hard to find a non Laser tach. Have been having problems with the new PT 99 from Monarch not rugged have had a lot of failures compared to older model. Think we will stop looking a live with the Monarch.
what should I do if the "ON" is blinking on the screen, I already tried to press the power button but nothing happened.
Asked by: minnie22
The meter has 3 modes which shut off the power differently so refer to the user's guide section 3 and 4 for turning off if not in measurement mode. If still a problem contact Omega's Temperature group at email@example.com.