Appelez-nous au 1-888-826-6342
Thermocouple surface Autocollant ou adhésivité du ciment, réponse très rapide <b>Vendu en paquets de cinq</b>Surface Thermocouple High or Low Temperatures! Self Adhesive or Cement-On! Super Fast Response All the Time!
Remises de volume disponibles
Calendrier de remise sur volume
Ajouté à votre panier
Veuillez ajouter au panier pour la conserver ou quitter la configuration
Veuillez ajouter au panier pour la conserver ou quitter la configuration
Échec de l'enregistrement des configurations, cliquez sur le bouton Configurer Recommencer
Délai de livraison (si pas en stock): 7 semaines
- PRIX MONTRÉS POUR DES PAQUETS DE 5
- Longueurs stockées de 1, 2, et 3 mètres
- Longueurs de fils sur mesure disponibles
- Disponible en thermocouples types K, T et J
- Installation simple - endossement autocollant au silicone [à 260°C ]
- Plage de température de -73C à 260°C continu, 315°C lors de l'utilisation avec le "ciment" (Ciments autosiccatifs OMEGABOND™ disponibles- placer le capteur et encapsuler avec ciment autosiccatif OMEGABOND™)
- Terminaisons à fils dénudés en standard (prise miniature moulée avec détente des contraintes intégrale en option)
- Dimensions du caoutchouc: 25 mm x 9,5 mm
- Egalement disponible avec surtresse acier inoxydable pour des applications plus résistantes
SA1XL-J-SRTC-SB Options de modèlesVoir tous les modèles
Modifiez ces spécifications pour commander un modèle différente. Toutes les combinaisons ne sont pas valides. Les options compatibles avec les sélections précédentes seront en gras.
*Les options en surbrillance ne sont pas compatibles. Veuillez sélectionner une combinaison différente.
Process Temperature Range
Termination Connection Type
Color Code Standard
DescriptionOMEGA présente la nouvelle génération de nos thermocouples à réponse rapide à montage en surface. Le nouveau thermocouple SA1XL atteint des temps de réponse de moins de 0,15 secondes. Ce thermocouple est construit en utilisant une jonction en soie de verre/Kapton avec un fil de sortie avec isolation en soie de verre. Le SA1XL peut être utilisé en tant que thermocouple auto-adhésif pour des températures jusqu'à 175°C ou il peut être cimenté sur place pour des températures jusqu'à 315°C.
Fournis en paquets de 5. Tous les modéles standard sont indiqués dans le tableau "Pour commander" ci-dessous. Consultez la notice technique (pdf) pour des informations concernant les longueurs non standard.
Rated 2 out of 5 by TestEngineer from Quality of product has dropped in recent years Having issues similar to those described by other users with units DOA. We have utilized this product for years (20-100 units/year) with little to no issues. We had been satisfied with the build quality and the ability to reuse these units several times if carefully applied/removed (5 stars), but within the last 1-2 years we have started to receive units that have been DOA (plug them in out of the box before even installing or removing adhesive backing, with no readout). The 5-packs that we are now receiving generally have 1-2 units DOA (you're lucky if this is discovered out of the box rather than after spending time installing it), not sure how this could be seen as acceptable, especially if a simple "These were received DOA" from the customer does not result in shipping of functional replacement units as long as the DOA units are returned. I have already sent back some of the bad units with a similar response from Omega indicating that a "full evaluation" would be performed. I received no additional information following the return of the units. Please Omega, revert back to whatever manufacturing process you were utilizing prior to these issues arising.
Date published: 2019-09-09
Rated 3 out of 5 by GPSKGF from Worked well - when they worked I bought 20 of these and they generally worked well. However, 4 of them did not function. The adhesive was good enough, but I wish it was stronger.
Date published: 2018-06-27
Rated 2 out of 5 by vacman from Something has changed...... The sample used to work great, had some robustness and never failed out of the box. in the last two years I've had to move away from using this product due to out of the box failures, the TC themselves becoming very fragile or early failures.
Date published: 2019-10-29
Rated 1 out of 5 by technician17 from ehh Received first set and 3 of 5 didn't work immediately straight out of packaging would get no reading on meter. other two proceeded to break and come apart soon there after. Received 5 probes in replacement very easily however one of these didn't work fresh out of the box also. wouldn't order these again.
Date published: 2018-06-01
Rated 2 out of 5 by Brandon from Too Fragile We have ordered several packs of these thermocouples and even taking the most care we can to not damage them during installation, at least 1 of 5 does not work after it is installed. There seem to be some that do not work right out of the box, even before the tape is removed from the adhesive strip (so apparently these came from the factory not working). This product is ideal for a lot of what we do but we are frustrated with the high rate of product failure before use.
Date published: 2018-04-02
Rated 5 out of 5 by Ian B from Good Thermocouple for Assorted Applications Durable and plenty accurate for our purposes. Self-adhesive makes for very convenient application to whatever surface you need to monitor.
Date published: 2023-09-25
Rated 5 out of 5 by LDavid from Surface TCs Exactly what we needed. Love that certificates are sent with packaging info.
Date published: 2023-08-14
Rated 5 out of 5 by Paul Labert from Every Omega TC we have used worked as intended. We have used Omega TCs and loggers in automotive R&D for over 20 yrs. Their TC is actually part of our Test Specification.
Date published: 2023-05-06
Referring to Omega Publication https://assets.omega.com/pdf/test-and-measurement-equipment/temperature/sensors/thermocouple-probes/tc_colorcodes.pdf What is the IEC tolerance class of the SA1XL-KI ? I want to know the accuracy.
Hi KM101 Thank you for the inquiry. The SA1XL-KI will follow the ANSI SLE and EN 60584-2 Tolerance Class 1. Best regards OMEGA
Date published: 2022-03-01
What is the accuracy of temperature measurement?
Thank you for your inquiry heasealina. SA1XL-E has an accuracy of +/- 1.0°C or 0.4%, whichever value is greater. For more information please review this link: https://assets.omega.com/pdf/test-and-measurement-equipment/temperature/sensors/thermocouple-probes/tc_colorcodes.pdf
Date published: 2022-08-12
We kindly want to verify SA1XL sensor: 1. the compatibility with the wireless receiver/transmitter known as "UW series compact wireless thermocouple/RTD transmitter"; and 2. can it be used several times on different surfaces?
Thank you for your inquiry. This sensor is compatible with the UW Series. These sensors are single-use. You can purchase TAP adhesive labels to re-attach sensors.
Date published: 2020-10-09
Ordering a quanity of 1 equals 5 complete thermocouples, Right?
Hi, you are correct. For this product, QTY 1 = a 5-Pack. Please reach out anytime if you have any further questions. Thank you.
Date published: 2019-04-26
It appears from the "ALL MODELS" list that the only type available with self-adhesive is a Type-T. All others (J, K, & E) are listed as "Cement On" only. Is that correct?
Thank you for your inquiry. All types have are self-adhesive. The J,K & E can be used with cement to go to elevated temperature. Type T cannot.
Date published: 2021-05-17
What is the gauge of the cable wire?
Thank you for your inquiry. The gauge of the cable wire is 30 AWG.
Date published: 2020-04-06
Is this thermocouple waterproof?
Thank you for your inquiry Conrad. No, this thermocouple is not waterproof.
Date published: 2022-11-07
Can these be configured with thicker Gauge wire or is the SB the only option for these?
Omega does not offer larger gauge wires for any of the SA1 series Thermocouples.
Date published: 2021-07-21